The settings from the first installation of the instrument are below the following ones taken in Dec 2019 after a reinstallation.
Screen snaps of key settings as of 27 Dec 2019 when spinning disk reinstalled.
The following table of laser powers was taken with the calibrated AOTF settings with Newport power meter 1928-C wand 818-ST SN 5110 taped to the 10X lens.
|405 nm||8.5 mW||4.5 mW||2.15 mW||0.86 mW||14.5 uW|
|488 nm||11.6 mW||5.8 mW||2.9 mW||1.17 mW||8.9 uW|
|561 nm||11 mW||5.6 mW||2.7 mW||1.0 mW||104 uW|
|640 nm||8.5 mW||4 mW||2 mW||0.8 mW||24 uW|
This could be changed to FAST.
Screen snaps of key settings as of 29 May 2019 when spinning disk installed.
Location of Bruker miniscanner calibration file: